The EM2010P/G and EM2010P/GA include everything you need to deploy the noise monitor and make unattended measurements for one week before recharging or replacing the battery pack.
The main steps for setting up are:
The measurements will start automatically and be uploaded to the web server as they become available.
The microphone comes fitted with weather protection (rain, snow, etc) and a wind shield. It mounts on a stainless steel post which in turn fits on the tripod.
A 10m cable connects the microphone back to the noise processor. This cable can be extended up to 30m using standard RG-174 coaxial cable and suitable connectors.
The batteries are contained in a separate carrying case. This is to make transport easier (batteries are heavy) and to give the option of multiple batteries to swap in the field without opening the noise processor case.
The battery pack connects to the processor case, and noise measurement will start automatically when the power is applied. With a full charge it has sufficient power to run the noise processor and mobile internet access for at least seven days.
Environmental Noise Monitor with GSM link and web access to measurements.